NOTICE: Our telephone provider is experiencing service interruptions. If you are not able to reach us by phone, please email email@example.com with your inquiry and we will get back to you as soon as possible.
Can We Help You?
Can We Help?
Dear Valued Stakeholder,
As we continue to monitor Coronavirus (COVID-19) developments closely, the health and well-being of our team, customers, and their employees is of utmost importance to us.
As a trusted vendor, we understand the imperative of limiting the impact this situation could have on our services. We are keenly focused on maintaining a safe work environment for our team while ensuring continuous service.
We have a robust integrated Business Resiliency Program in place and are committed to keeping our operations running smoothly.
This Plan includes:
Minimizing supply chain disruptions through constant communications with our production facilities and logistics partners
Maintaining larger stock levels of products at our distribution facility
Prioritizing orders being shipped based on the order in which they were received
Enabling work from home capabilities for our sales and support staff
Providing our team members with information and best practices to prevent the spread of any illness
Coordinating communications with our team, associates, customers and partners
Limiting all non-essential business travel
In the short term, you might experience a slightly longer than anticipated lead time for fulfilment of some the orders. Please be assured, we are taking every measure to ensure minimal disruptions or delays and continue to monitor this fluid situation on a daily basis.
Thank you for your business, and your continued support.
Dimitry Fedorov, President
Laser Beam Profiles
Laser Beam Profiles
A TEM00 beam has an intensity profile that is very nearly Gaussian (>95%) and does not change as it propagates (i.e. from when it emerges from the laser to the far field). It is desirable to have a laser operate in this mode because the closer your laser is to a perfect Gaussian shape, the tighter it can be focused and the lower the divergence of the beam.
M2 is a measure of how close a laser beam's intensity profile is to a perfect Gaussian distribution. If it were a perfect Gaussian beam, then M2 = 1. For any other laser beam, M2 > 1. Depending on the application and industry, M2 values of 1.1, 1.2, and even up to 1.5 may be considered high quality TEM00 beams.
Intensity profile of a TEM00 beam from a Laserglow 532 nm DPSS laser. This measurement was made using a CMOS beam profiling camera at Laserglow's testing facility.
Graph of the intensity profile of the TEM00 beam measured in figure 1 is shown in black. A perfect Gaussian intensity profile is overlaid in blue for comparison.
Near TEM00 Beam
All single Transverse Electric Mode diode lasers are considered near TEM00 or Quasi Gaussian. Most have a beam that can be accurately approximated by a Gaussian distribution, but with a slightly narrower central maximum and longer tail.
Because of the structure of laser diodes, the divergence of the beam is greater along one axis (known as the fast axis) leading to an elliptical beam shape. They also have a highly divergent beam compared to other types of lasers, and as a result need to be collimated with additional optics.
Greater divergence along the fast axis causes the beam to be clipped by the edges of the lens, which creates a diffraction pattern that manifests as side peaks along the edges or twin peaks in the center of the beam. This diffraction pattern changes shape as the beam propagates. For a collimated beam, the diffraction pattern will eventually disappear in the far field, leaving a quasi Gaussian profile.
Alternatively, diffractive effects can be removed in the near field using a technique known as spatial filtering. The M2 of a near TEM00 beam can range from 1.1-1.7. The majority of near TEM00 beam diode lasers offered by Laserglow are listed with an M2 < 1.5.
Intensity Profile of a Near TEM00 beam from a Laserglow 405 nm Diode Laser. The beam is elliptical in shape with some diffractive peaks along the edges caused by clipping at the collimating lens, but can still be seen to have a nearly Gaussian profile.
A graph of the intensity profile measured in figure 3 is shown in black. A perfect Gaussian intensity profile is overlaid in blue for comparison. Despite some diffractive peaks, we can see that the intensity distribution is nearly Gaussian, and is regarded as Near TEM00.
Multimode Laser Diode Beam
To achieve higher powers using laser diodes it is necessary to increase the width of the diode's active layer. Such laser diodes are often called wide stripe laser diodes or broad area laser diodes. The beam emitted from a wide stripe active layer contains multiple TE modes. Each of these TE modes is quasi-Gaussian near TEM00 which combine to form a multi-TE mode beam.
As the beam propagates, the individual modes gradually increase in size until they merge together to form what appears to be one wide central maxima in the slow axis direction. This gives the beam a rectangular looking shape. However, when the beam is focussed, the intensity profile of the focussed spot will appear to be several discrete peaks like that shown in figure 5a.
Compared to TEM00 mode beams, multimode beams cannot be as well collimated or focussed to as small of a spot size. The M2 of multimode beams can vary greatly from around 3-4 for a multimode laser diode to greater than 20 for certain types of high powered lasers.
Intensity Profile of a Multimode beam from a Laserglow 447 nm Multimode Diode Laser. The beam appears rectangular and can be seen to have 3 central maxima along the slow axis. The beam also exhibits some diffractive rings along the edge of the fast axis caused by clipping at the collimating lens.
A graph of the intensity profile of the Multimode Laser Diode beam measured in figure 6 is shown in black. A perfect Gaussian intensity profile is overlaid in blue for comparison. Three intensity peaks are clearly visible along the slow axis. Each peak is an individual TE mode.